The pen- type visual fault locator is specially designed for field personnel who need an efficient and economical tool for fiber tracing, fiber routing and continuity checking in optical networks. And it is the essential test tool for the FTTH project. This visual fault locator is with so many advantages, such as long woking life, rugged, portable, beautiful appearance and so on. It is the best choice for the field personnel.
1.2.5mm universal connector,for 1.25mm connectors,FC(male)-LC(Female) adaptor also can be provided on request
2.Operates either in CW or Pulsed mode with constant output power
3.Low battery warning
4.Long battery life (up to 60 hours)
5.Drop-resistant and dust-proof design of laser head
6.Laser case ground design prevents ESD damage
7.Portable and rugged
8.Easy to use
1.Find breakpoints, poor connections, bending or cracking in fiber optic cables
2.Find faults in an OTDR dead zone
3.Is used for end-to-end visual fiber identification.
1.Maintenance in telecom, CATV
2.Test Lab of optical fibers
3.Fiber routing and continuity checking in optical networks
4.Other fiber optic measurements
650nm±10nm (635nm is available on request)
2.5mm universal connector, for 1.25mm connectors, FC (Male)-LC. (Female) convertor can be provided on request
Both CW and Pulse available
2Hz to 3Hz / 9Hz
2AA alkaline batteries
LR6 AA alkaline battery
-20°C to 60°C
-40°C to 85°C
106g (without battery)
Method of use
1. Screw the end cap from the pen, the battery will be prepared into the pen, battery cathode to the end cap, the positive on the pen body is positive;(note: please do not turn pen and pencil tail into the battery) 2. Put the pen and pencil body connection installed, open the dust cap, press the control switch one time, observe the interface has a red light, luminous tube on at the same time;(note don't eyes were light interface, so as not to damage the eye) 3. Press the control switch again, observe the light into a pulse mode, the indicator light-emitting tube and out of sync pulse light;(pulse frequency in 2-3 hz or 9 hz) 4. Press the control switch again, light source closed, no light output, at the same time indicator light pipe out;(switch mode to: continuous - pulse - closed - continuous cycle mode) 5. When detecting insert tested optical fiber optical interface, press the control switch at the same time, choose the output light work mode (continuous or pulse); 6. After use, build the dust cap, do not use for a long time, please take out the battery, in order to avoid battery rot, damage the light source.
Maintenance and matters needing attention
1. Laser and harmful, especially to pay attention to protect the eyes, when laser works, avoid to irradiate the eyes with laser directly;
2. In generally, the higher the temperature, the shorter the life of laser, when used to avoid high temperature working environment; 3. The work of laser optical fiber head should be cleaned; 4. When used, the optical interface and the interface needs to detect light and light pull plug; 5. Don't use the light source, would you please put dust cap to avoid dust fall; 6. Should remove the battery when not in using; 7. Please do not remove the end cap and any part of the dust cap, remove shall not be responsible for damage caused.
Bob LASER CO., LTD is a factory specialized in laser sight and a series of laser products. Our products has safe quality and competitive price and complete after-sale service system .